|Published (Last):||7 February 2008|
|PDF File Size:||15.26 Mb|
|ePub File Size:||16.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
So, basically DIN is not anymore used with the exception of revision But there is also a technical reason why DIN should not be used anymore. DIN IEC interface leakpoint leakfree flangeconnection leakage manifoldinterface manifoldtotransmitter flangeconnection transmitterinterface.
Jump to. Sections of this page. Accessibility Help. Email or Phone Password Forgot account? Sign Up. KG on Facebook. Log In. Forgot account? Not Now. Recent Post by Page. The expectation of the users is to get an accurate result of the pressure measurement. But long gauge lines will definitely cause inaccurate measurements because the lines dampen the signal read at the pressure transmitter.
Liquids and condensate in long gauge lines can potentially freeze causing measurement error. More reliability - even if the going gets tough. Those who opt for p For example, more reliability: Because even in the event of a fire, our products are absolutely reliable — Fire Safe. Do you know the most compact and efficient way to perform a pressure We do! Simply use Monoflanges.
MX26 2-Valve Manifold